GAS201L Total Sulphides Gastec Gas Detector Tubes 0.002 to 0.02 mg
GAS201L Sulphides Gastec Detector Tubes
1. The measured out mud and soil is put in the generation tube and 5mL of distilled water are added.
In case of muddy water, it suffices to add a small amount of distilled water (measured out in a graduated cylinder) to the gas generation tube.
Put the cap on the generation tube.
4. Pull the handle of the gas sampling pump to draw in a sample of the generated hydrogen sulphide.
(By repeating this process several times, all sulphides become hydrogen sulphide several times repeatedly).
The scale in the detector tube is read, and the concentration of the sulphide in the sample is calculated.
|Measuring Range||0.002 to 0.02 mg|
|Sampling Time||45 seconds per pump stroke|
|Detecting Limit||0.0002 mg (n=1)|
|Colour Change||White Brown|
|Reaction Principle||Gus generation tube: Sulphides + H2SO4 H2S
Detector tube: H2S + Pb(CH3COO)2 PbS
|Coefficient of Variation||5% (for 0.002 to 0.02mg)|
|Shelf Life||3 Years|
temperature & humidity
|Store the tubes at cool and dark place.|
|Possible coexisting substances and their interferences|
|Substance||Concentration||Interference||Change colour by itself|
|Chloride ion||–||No error||No discoloration|
|Chromate ion||–||No error||No discoloration|
|Carbonate ion||–||No error||No discoloration|
|Calibration gas generation||Static gas dilution method|
Talk with the Experts
Contact us - Our application specialists will identify the best product for your needs based on the information that you can provide. If you're unsure about what's important, consider booking a site survey.
Book a site survey - Our specialists consider your process, your site conditions and work with you to understand your needs. It is the most effective means of ensuring your system is completely fit for purpose.