GET A FREE DESIGN CONSULTATION
NEXT WORKING DAY DELIVERY when you order before 2pm
REQUEST A CREDIT ACCOUNT
GAS81D Acetic Acid Passive Dosi-Tubes 0.5ppm – 100ppm
GAS81D Acetic Acid Gastec Passive Sampling Dosi Tubes (Time Weighted Average Detector Tubes)
The Dosi-Tube System allows an operator to evaluate the level of gas a worker is exposed to over a time period such as a working shift.
The Dosi-Tube is easily clipped into the Dosi-tube holder (GAS710) which is worn on a garment of clothing.
|Measuring Range||0.5 to 100 ppm|
|Sampling Hours||1 to 10 hours|
|Detecting Limit||0.2 ppm (10 hours)|
|Colour Change||Purple Yellow|
|Reaction Principle||Acetic acid neutralizes alkali to discolour the indicator to yellow.|
|Coefficient of Variation||5% (for 5 to 20 ppm∙hr), 10% (for 20 to 100 ppm∙hr)|
|Shelf Life||Up to 3 years from point of manufacture|
temperature & humidity
|Temperature correction is necessary|
|Store the tubes at cool and dark place.|
|Possible coexisting substances and their interferences|
|Substance||Interference||Change colour by itself|
|Hydrogen Chloride, Nitric Acid||Plus error||Produce yellow stain|
|Chlorine, Nitrogen Dioxide||Plus error||Produce yellow stain|
|Hydrogen Cyanide, Sulphur Dioxide||Plus error||Produce yellow stain|
|Other substance measurable with this detector tube|
|Substance||Correction||Sampling Hours||Measuring range|
|Formic Acid||Factor: 1.1||1 to 10||0.55 to 110 ppm|
|Acetic Anhydride||Factor: 0.6||1 to 10||0.3 to 60 ppm|
|Calibration gas generation||Diffusion tube method|
|Multi Gas Tags||
Acetic Acid (CH3CO2H), Acetic Anhydride ((CH3CO)2O), Formic Acid (HCO2H)
Talk with the Experts
Contact us - Our application specialists will identify the best gas monitoring equipment for your needs based on the information that you can provide. If you're unsure about what's important, consider booking a site survey.
Book a site survey - Our specialists consider your process, your site conditions and work with you to understand your needs. It is the most effective means of ensuring your system is completely fit for purpose.